HMO2524 / 352x: Correct connection of the probe for adjustment (ID #550)HMO2524 / 352X: How to connect a probe correctly for adjustment
In order to do a correct LF and HF adjustment for a passive probe with the HMO scope series, you first need to contact the probe to the probe adjust output. Please move the probe tip into the hole of the right pin of the probe adjust output under the display of the HMO Scope. Then move the probe in a way, that you make a good mechanical and electrical connection of the ground ring (metal ring above the probe tip) with the left pin (Ground) of the probe adjust output. For refeence please see the drawing.